Deliver growth-boosting CO2with our user-friendly, easy to operate system. With its programmable timer, it will automatically maintain ideal CO2levels for optimum growth. For small to medium size areas, select the COSYS; for larger than 10' x 10' areas, select the COSYS20.
To use Hydrofarm'sCO2Flow Rate Calculator, clickHERE
NOTE: Current COSYS & COSYS20 now come equipped with a black rubber O-ring that has replaced the white plastic
Product DetailsAutopilot APC8200CO2Monitor & Controller with Memory offers a simple and affordable way to control, monitor and recordCO2levels over an adjustable period of time.Trace your room'sCO2levels over minutes, hours, or up to 7 days with this easy-to-use controller. The APC8200 also boasts a fully customizableCO2setpoint and deadband. Adjust and review yourCO2levels when you want to and how you want to. The APC8200's memory featuregives growers a real edge in fine-tuning the grow room environment. Optimize yourCO2levels with this precision controller from Autopilot!FEATURES:
- Fully customizable ppm deadband and CO2setpoint allow flexibility in programming
- Trend Chart records CO2levels per minute, hour, day and week for accurate analysis of CO2distribution
- Adjustable Trend Chart Zoom levels let you trace your ppm levels
- Remote CO2sensor probe w/photocell & 15' data cable lets you take readings from proper grow room locations
- 2-Channel Low Drift NDIR Sensorprecisely reads CO2ppm levels
- Green LED identification lights inform you of the unit's status at all times
APC8200 Error Codes:
Fault IconDescription of the faultSuggested ActionsErr4IR lamp errorPlease reconnect power adapter, if the Err4 still appears, please contact the service department.Err5Internal parameter errorPlease reconnect power adapter, if the Err5 still appears, please contact the service department.Err6Sensor unit Connect ErrorPlease reconnect sensor unit and redo power on, if the Err6 still appears, please contact the service department.Err7temperature or humidity errorPlease turn off power adapter, and put the device in a dry place more than 12 hours.Err8Internal calculate errorPlease reconnect power adapter, if the Err6 still appears, please contact the service department.